My-B015 Vet Urine Analyzer
|FOB Price:||US $350-400 / Piece|
|Min. Order:||1 Piece|
|Min. Order||FOB Price|
|1 Piece||US $350-400/ Piece|
|Transport Package:||Standard Exports Package|
|Payment Terms:||T/T, Money Gram, Western Union|
- Model NO.: MY-B015
- Recording Method: Thermal Printer / External Pin Print / LCD Screen
- Data Storage Volume: 2000 Copies of a Single Laboratory
- Trademark: Maya
- Origin: China
- Test Speed: 60 Copies of Test / Hour; Continuous Test of 120
- Recording Paper: 57mm Thermal / Non Thermal Paper
- Power: AC 220V ± 10%, 50-60Hz
- Specification: CE
8 items, 10 items and 11 items test strips are for choice;
Thermal printer or needle printer is for choice;
Color and clearance of urine are available;
13 items are also available (Creatine and Calcium included)
1-Principle of Determination: principle of determination by ultra-high degree of cold light reflection
2- The wavelength of monochromatic light: 525nm, 610nm, 660nm;
3-Product determination: glucose (GLU), bilirubin (BIL), ketones (KET), specific gravity (SG), pH (pH), occult blood (BLD), protein (PRO), urobilinogen (URO), Nitrite (NIT), white blood cells (LEU), ascorbic acid (VC), creatinine (CRE), calcium (CAL) and albumin (MAL).
4-Strips type: Manufacturers Product can support following urine test strips: URS-8, URS-10, URS-11, URS-12, URS-10E, URS-10N, URS-11E, URS-11N, URS-11EA, URS-12E, URS -12F.
5-Test speed: 60 copies of test / hour; continuous test of 120 / h.
6- Recording Method: thermal printer / external pin print / LCD screen display.
7- Recording Paper: 57mm thermal / non thermal paper.
8- Data storage volume: 2000 copies of a single laboratory
9- Ideal environment: the ideal environment for instrumentation temperature range from 15 °Cto 35 °C, humidity PH ≤ 80%.
10- Temperature of the strip of paper in the ideal environment for measuring:range 20 °C ~ 28 °C
11-Data communication interface: the serial port.
Transfer rate: 4800bps, stop bit: 1 bit, Parity: None, start bit: 1
12-Power: AC 220V ± 10%, 50-60Hz
13-Power: 50VA; fuse model: φ5 × 20,250 V/3.15A
14-NW ≤ 3Kg
15-Working structure and Principe: During the test the strip reagent zone and the biochemical composition of urine react together to produce a change of color. The deep or shallow change of the color will receive irradiation from the monochromatic light and then will reflect as different rates. The content of biochemical substances in urine can thus be determine according to the gradient of reflectivity and the test area on a number of reagents, this can be used to realize a number of biochemical substances in urine test. The Instrument operate by the scan of the test paper by three different kind of monochromatic light , after scan the system will convert optical signals into electrical signals, the micro-controller receives the electrical signals and then go through the procedure of handling, processing, and calculates a color reflectance values.