My-I017b Maya Medical Electric Parturition Bed with Good Price
|FOB Price:||Get Latest Price|
|Min. Order:||1 Piece|
|Payment Terms:||T/T, Money Gram, Western Union|
- Model NO.: MY-I017B
- Folded: Folded
- Condition: New
- Width: 550mm
- Min.Height: 480mm
- Origin: China
- Type: Electric Bed
- Certification: CE, SGS
- Overall Length: 1200mm
- Max.Height: 1200mm
- Trademark: MAYA
Electric Parturition Bed(Model MY-I017B)
a) Usage:This bed is designed for women's parturition and gynecological operation.
b) Description: The bed surface is streamline and conforms to the patient body posture.The dynamoelectric adjustment satisfy whole inclination.The obstertric table altitude is adjusted by the whole inclination.It can be especically used for gynecology,urology surgery,mirror check and diagnosis.
Dimensions of Table Top:
Table Top from floor:
Range of movements:
Device safety sort:I B
3,Parturition bed composition:
The parturition bed is composed of the mechanism and the electrical control system.
The mechanism is composed of the bed body, the leg frame, the contaminative basin, the auxiliary equipment and so on, the electric appliance control system is composed of inside controller, the transmission system, handle controller and so on.
1)The dynamoelectric adjustment satisfy whole inclination.The obstertric table altitude is adjusted by the whole inclination. It has a moving controlling push button container attached. The controlling push button are all dot buttons, when the necessary button is rushed, the corresponding motor will work, when the button is not puessde, all movements stop.
2)The bed surface,the head pillow is made of polyurethane cushion which is easy to unlade and sterilize.
3)There is a hidden dirt basin below bed pots, which can be easily pulled out or pushed in to facilitate treating dirty water and debris.
4) The moving and stabilization of the base. By loosening Handle of the base clockwise, the table can move. On the contrary, it can be stabilized.
5)Leg rack can be moved front and back, around rotation, height adjustment, Lock convenience.
6) The Company will be able to provide circuit draws in accordance with users' needs.
1) Lubricating oil should be added to every active part, so that the bed can be operated easily and prolong their life. Smear Vaseline over the tracks of the accessories on each side of the bed and the racks under the table, thus preventing it from being rusty.
2) Before putting into use, all attachments should be checked to see if they are locked securely so as to prevent them from loosing or sliding off while used.
3) keep the bed and accessories clean and prolong their life.
4) The ground must be kept dry. While the floor is washed and scrubed, please try not to make water spill into the plate or on the rotated wheels, thus preventing it from being rusty.
5 )Put the bed in dry place to avoid the electric controlling devices being wet and being damaged.
6) Keep the button container dry and clean to avoid current leakage. Keep it in the proper
place after being used.
7) The product can electrically operate 2 movements. These movements must be operated separately.Attention: the product is supplied by single-phase alternating current (220V, 50Hz), tri-socket .Fuse plug rating is 2A .Please carefully inspect socket to insure reliable earth; shut off power while stop perform. When trouble happen please inform company /agency to assign engineer overhaul, don't overhaul by yourself. Don't press cables avoiding to damage cable or isolation during usage.
8)It must be repaired and kept clean regularly once a year.
9) The replacement parts during Electric parturition bed normal using period, has been confirmed scrapped (unable to use), should be handled according to the classification (recyclable and non-recovery).
10) The service life for Electric parturition bed is seven years. If it is used in expired period, it maybe led to wearing and corrosion of some mechanical parts, and had a negative influence upon the mechanical intensity. The patients maybe suffered. So the operating table should be used in the regular maintenance and during safety using period.
11) Electric parturition bed must be used on the antistatic ground.
Warning: electric parturition bed coupled with high-frequency surgical equipment for use, it is possible for the operator and patients to be burned.
7,Cleaning, disinfection and sterilization
1,Washing the operating table with water is prohibited.
2,Clean operating table with a soft cloth, sponge, wet cloth or detergent. Not use a hard brush, steel wire brush, decontamination powder, paint, ether, bananas water, acid-base solution to any surface.
3,Mattress must be disinfection with alcohol, ether, ultraviolet disinfection and so on (abandoning high-temperature disinfection).
Transportation and stockpile environmental conditions
Moving environmental conditions
c)Atmospheric pressure: 70.0kPa~106.0kPa.
d)Supply voltage: AC220V,50Hz±1 Hz
This device in package should be stored indoor with a relative humidity less than 80%, no erosive air and good ventilation is required. No side-sloping.
It can be shipped in regular transportation without crash, oscillation, rain or direct sunlight.