My-G032f Hot Selling Bluetooth Forehead Thermometer
|FOB Price:||US $12-20 / Piece|
|Min. Order:||1 Piece|
|Min. Order||FOB Price|
|1 Piece||US $12-20/ Piece|
|Production Capacity:||500 Set/Sets Per Month|
|Payment Terms:||L/C, T/T, Western Union, Money Gram|
- Model NO.: MY-G032F
- Quantity Per Carton: 60PCS
- Trademark: MAYA
- Specification: CE
- Retail Packing: Colorful Box
- Carton Size: 58.3*40.2*35cm
- Transport Package: Carton
- Origin: China
Retail Packing: Colorful Box
Quantity Per Carton 60PCS
Carton Size 58.3*40.2*35CM
Gross Weight 15kg
Net Weight 14kg
volume weight 18kg
This product is a professional non-contact infrared thermometer with bluetooth function(bluetooth version is 4.0),which can be used to measure human body and forehead temperature .
Measure at different part of body may result in different reading as because the more body exposed the more severe body temperature will be impacted by environment.
2. User safety manual
·Please read this manual carefully before use.
·The application environment temperature range for this product is from 10ºC to 40ºC, the best usage environment temperature is 25ºC.
·Please do not use this product in the environment with temperature higher than 50ºC or lower than 0ºC.
·Please do not contact this product to any electrified device as to avoid electrical shock.
·Please do not use this product in environment of humidity higher than 80%.
·Please do not approach this product too close to electromagnetic emitting device such as radio or mobile phone.
·Please do not expose this product direct under sun, or too close to oven and no contact with water.
·Please do not hit or drop this product, stop using it if it is damaged.
·The sweat, hair, hat or scarf may impact the measurement accuracy.
·Please make sure the measurement distance is within 5 cm.
·In order to get stable and reliable measurement result, should keep this instrument inside the room for 15 to 20 minutes before using it if the room temperature changes too much.
·Please measure behind the earlap if the forehead can not reflect the actual body temperature because of sweating.
·Use alcohol to sweep the surface of the instrument gently for cleaning.
·Please contact the retailer for reparation instead of doing it yourself.
·High precision non-contact body temperature measurement.
·Can choose ºC or ºF
Temperature mode selection, Press "MODE" when instrument in "ON" status.
·Body inner and body surface temperature measurement mode,( Body inner mode is for body inner temperature, Body surface mode is for measuring object surface temperature)
Setting : Hold the "MODE" key for 4 seconds till "F-1" appears, and then press upwards and downwards key for shifting.(BODY - body inner temperature; Surf-body surface temperature)
·Adjustable alarm temperature (the preset alarm temperature is 38 ºC for this product)
Setting: Hold "MODE" for 4 seconds till "F-1"appears, press "MODE"to enter "F-2"(preset value is 38), and then press upwards and downwards key to change the preset value to what you want, and then press "MODE" three times to turn off the instrument, setting completed.
·Buzzing alarm function (Buzzer can be set to on or off)
Setting: Hold "MODE" key for 4 seconds till "F-1" appears, and then press "MODE" for another 3 times to enter "F-4" (the preset value is "ON"), and press upwards and downwards key to choose (On-Turn on, Off-turn off),and then press "MODE" to quick the setting menu.
·The LCD with backlight is suitable the usage in weak illumination.
Setting: Hold upwards or downwards key for 5 second to turn on the backlight.
·Automatic adaptable range; resolution is 0.1ºC(0.1ºF).
·Can store the most recent 32 measured data (press upwards and downwards key can access to the stored data)
·Automatic data storage and shut down.
·Bluetooth version : Bluetooth 4.0
·Bluetooth transmission function : Can be connected with cell phones (iPhone 5) or computers
(iPad) and other higher instruments ,so that it will be convenient for data's storage and analysis .
In the boot condition , when "∑" sign is flashing in the lower left corner of the screen , it shows that the bluetooth thermometer has entered the matching and searching condition.
If the pairing is successful, the "∑" sign will not flash in the lower left corner of the screen, it shows that the bluetooth thermometer is paired with the instrument successfully .
4.Additional direction for usage
Non-contact infra-red thermal meter is a kind of professional body temperature instrument by measuring the forehead, has been widely adapted for general application, the product comply ASTM1965-1998 standards, this product can not take the place of the diagnoses from doctor.
5.Notice before usage
In order to get stable and reliable measurement data, the instrument should be kept in room before usage if room temperature has big change. The following necessary inspection should be finished before usage.
First step: measure body temperature with traditional body temperature meter, assume 37.5ºC(99.5ºF ) is the reading.
Second step: use none-contact thermal meter to measure the same person who has been measured the body temperature mentioned in first step, hold the meter 5 cm away from the forehead (pay attention to move away any obstacle such as hair or sweat) , if the same reading 37.5ºC(99.5ºF) is obtained, then the meter is good for use.
If lower reading is obtained, such as 36.4ºC (97.5ºF), and 1.1ºC(2.0ºF)in difference, calibration is needed. Hold "MODE" key after turning on the instrument for 4 seconds, screen appears "F-1", then press "MODE" for two times and enter "F-3", and then press upwards key to increase the offset and downwards key to decrease the offset, and then press "MODE" two times to turn off the instrument, calibration is completed.
Third step, follow the second step again to confirm the instrument is good to use.
."MODE" selection key
7.LCD screen description
.Human body temperature mode (BODY)
.Surface temperature mode (SURF)
.Storage data withdraw
.Bluetooth Pairing sign: "∑"
8. technical indexes
|Power||DC 3V(two #7 battery in series)|
|Weight||Gross 267g, Net 137|
|Measurement range in human body mode||32.0-42.5ºC|
|Measurement range in surface mode||0~100ºC|
|Precision in human body mode||±0.3ºC(0.54ºF)|
|Precision in surface mode||±1.0ºC(1.8ºF)|
|Measurement distance range||5~15cm (2~5.9 inch)( Best measurement distance 5 cm)|
|Time to automatic shut down||15 seconds|
None-contact infra-red forehead thermal meter precision
|32~35.9ºC(89.6~96.6ºF)||±0.3ºC(0.5ºF)||According to ASTM standards E1965-1998(2003)|
9-1. Battery installation (pay attention to the positive, negative polarity)
9-2. First time usage or with new battery, wait 10 minutes for warming up. If the meter has been shut down for a long time, the meter will perform environment temperature check after turning on, and ready time will be 1 to 2 seconds longer.
9-3. Point to the forehead at a distance of 5 cm, press measurement switch, temperature reading appears immediately. Make sure the forehead is not covered by hair, sweat, cosmetic or hat.
9-4. When sweating or environment temperature changes too much as to affect the forehead temperature, point the meter to the position behind the earlap, and make sure that body surface is not covered by hair, sweat, cosmetic or hat.
9-5. When press down the measurement switch , the screen appears the reading of the temperature under measurement, the measurement data can be stored automatically( press upwards and downwards can access to the most recently stored 32 data)
In the boot condition , when "∑" sign is flashing in the lower left corner of the
screen , it shows that the bluetooth thermometer has entered the matching and
9-7.If the pairing is successful, the "∑" sign will not flash in the lower left corner of the
9-8.Bluetooth transmission function
Can be connected with cell phones (iPhone 5) or computers (iPad) and
other higher instruments ,so that it will be convenient for data's
storage and analysis .
9-9. Change battery
When it is out of battery, a signal will appear on the screen as a reminder to change battery, open the battery cover and pay attention to the polarity of the battery, wrong polarity could lead to instrument damage.
.Please take out the battery for long time idles, in case the battery liquid leak out and damage the instrument.
9-10. Usage life
The usage life is about 40 thousand times of measurement.
As to obtain a stable and reliable measurement data, the instrument should be kept in room for 15 to 20 minutes if the change of room temperature is big. Human body temperature could change during the day, and at the same time other element could affect the body temperature such as age, agenda and skin color.
Normal temperature range for different part of body
|Body part||MethodNormal TempºC||Normal TEMP ºF|
Normal temperature range for different age
·Protection glass of the LCD is very important and fragile, treat it carefully during usage.
·Please do not attempt to charge the battery which is not chargeable, do not throw the battery into the fire.
·Please do not expose this product directly into the sun, keep it away from the water.
A,·The detection head is the most sophisticated part, treat it with care.
·Clean the instrument surface with wiper and 70% density alcohol.
·Please do not use corrosive cleaning agent.
·Keep it away from all the liquid.
·Please store this product in dry environment free of dust, contamination and direct sun.
B. If the following problem appears during usage, please follow the maintenance instruction. If the problem insists, please contact the customer service.
.LCD can not display value reading.
In body temperature mode, if body temperature is lower than 32ºC or higher than 42.5ºC, "LO" or "HI" will be displayed instead of value.
. "HI" message on LCD
When "HI" appears, this signal indicates that the temperature is out of the measurement up limit or the body temperature is higher than 42.5 ºC for in body mode.
. "LO" message on LCD
When "LO" appears, this signal indicates that the temperature is out of measurement range low limit or body temperature is lower than 32ºC for in body mode.
The possible situation for reference when "LO" or "HI" appears
|The root cause for "LO" or "HI"||suggestion|
|Temperature reading is impacted by hair or sweat||Make sure the measurement body part is not covered by any obstacles.|
|Temperature is affected by airflow||Make sure the airflow is stable during measurement|
|Measurement distance is too big||Please keep the measurement distance smaller than 5 cm|
|Move from low temperature or high temperature outside environment into the room||Wait for 10 minutes till the body temperature adapt to the new environment.|
electromagnetic compatibility declaration
This product has been tested and certified the electromagnetic compatibility according the EU standards 60601-1-2:2007, but it is not guaranteed that this product is absolutely resistant to electromagnetic interference, please avoid to use this product in the environment of high electromagnetic.
For electromagnetic compatibility, medical electrical instrument needs special attention and should be installed and used according to the electromagnetic compatibility information provided in the attached documents.